Could you/anybody explain precisely what is happening in the INST mode?
The only thing I can find about it is in the users manual:
"Finally, if CPS > 255, we report CPS*60 and switch to INST mode, since we can’t store data in
the (8-bit) sample buffer."
And that doesn't shed much light on it.
Support forum for the mightyohm.com Geiger Counter.
2 posts • Page 1 of 1
Who is online
Users browsing this forum: No registered users and 0 guests